Sepax kits are integral to our Sepax Cell Separation System. They are supplied sterile for single-use and enable a functionally-closed environment to separate cellular product into its components.
The CS-530.1 kit is designed for processing umbilical cord blood (UCB) in order to concentrate the nucleated cell fraction into a single-compartment CryoSC-S cryobag.
|1||Separation Chamber||220 mL spinning chamber enabling the centrifugation and transfer of processed product|
|0.2 µm filter that connects to the Sepax pressure sensor|
|Three automatically controlled rotational valves to direct fluid flow|
|4||Drip Chamber||200 μm filter to block blood clots and trap air bubbles|
|5||Initial Line||Connects the kit to the initial bag via a spike or SCD connection|
|6||Cryobag||CryoSC-S cryobag with 10 to 30 mL storage capacity|
|7||By-product Bag||500 mL PVC by-product bag to collect plasma|
|8||DMSO Extension Line||A line equipped with leur port connections for adding cryoprotectant solution|
|9, 10||Final line||Final line with luer and sampling ports|
Kit / protocol compatibility matrix
|Kit / protocol||UCB||UCB-HES|
Environmental conditions and compliance
Store kits in a clean and dry environment without chemical or biological contamination within the following conditions:
|Mode||In operation||Storage and transport|
|Temperature||+7 to +27°C||+4 to +40°C|
|Relative humidity||30 to 75%, non-condensing||20 to 75%, non-condensing|
Shelf life: Two years from the date of manufacture.
Sterilization: Ethylene oxide.
Material: Standard medical grade PVC, PP, EVA, and other materials.
Biocompatibility: ISO 10993-1
Fluid path: Non-pyrogenic
The CS-530.1 is CE marked and complies with the applicable requirements of the 93/42/EEC Medical Device Directive. Confirm with your local regulatory authority for compliance with other regulations.
CS-530.1: Product #10015
Packaging: Individual blisters sealed with a Tyvek® cover.
Minimum order: Box of six kits (dimensions: 33 x 35 x 26 cm, 13 x 14 x 10 in. ).